75x55mm Manual Stage
|XY Mechanical Stage||75x55mm|
|XY Stage Travel Distance||Manual|
|XY-Axis Drive Mode||180x155mm|
|Stage Backlight Window Size||40x90mm|
|Stage Scale||X: 100-180mm Y: 10-70mm|
|Stage Knob with Scale||With Scale|
|Number of Stage Clips||1 Pair|
|Surface Treatment||Electroplating Black|
|Net Weight||1.54kg (3.40lbs)|
|Packaging Type||Carton Packaging|
|Packaging Material||Cardboard Box|
|Packaging Dimensions(1)||21x19x9cm (8.268×7.480×3.543″)|
|Inner Packing Material||Plastic Bag|
|Ancillary Packaging Materials||Styrofoam|
|Gross Weight||1.70kg (3.75lbs)|
|Minimum Packaging Quantity||1pc|
|Transportation Carton||Carton Packaging|
|Transportation Carton Material||Cardboard Box|
|Transportation Carton Dimensions(1)||21x19x9cm (8.268×7.480×3.543″)|
|Total Gross Weight of Transportation(kilogram)||1.71|
|Total Gross Weight of Transportation(pound)||3.77|
Microscope stage is usually located under the objective lens of the microscope to place the object to be observed. Usually, it is equipped with mechanical motion devices, and is moved and positioned in XYZ three directions or rotates or tilts around the Z axis and other functions.
Microscope stage usually has a function of moving in the XY horizontal direction, and is required to be perpendicular to the optical axis of the microscope Z direction in the XY horizontal direction.
XY Mechanical Stage
There are motion and positioning mechanisms in the XY horizontal direction. In general, the XY stage needs to adjust the level of the stage, so that the stage plane and the main optical axis of the microscope Z direction are perpendicular.
Stage Backlight Window Size
Stage backlight window size refers to the size of the window through which the transmitted light passes under the stage on the XY table plane of the stage. This window is usually covered with a piece of glass. For some stages with accuracy requirements in the XY horizontal direction, the horizontal plane of the glass can be adjusted by the height of the screws on the four corners below, and the consistency with the height of the stage plane is guaranteed.
The movement of the microscope stage or the mechanical stage can be measured by the moving distance of the ruler, and the size and area of the sample details can be calculated. The ruler can be divided into main scale and sub-scale. The minimum grid value of the main scale is 1 mm, the integer is measured; the minimum grid value of the sub-scale is 0.9 mm, the decimal is measured. When measuring, if what the main ruler measures is not an integer and therefore one needs to read the decimal of the specimen, align the end point of the sub-scale to the end of this specimen, and then find the scale on the line of main ruler and the sub-ruler, and see which group is the closest, the length of this decimal is the reading of the sub-scale.
After unpacking, carefully inspect the various random accessories and parts in the package to avoid omissions. In order to save space and ensure safety of components, some components will be placed outside the inner packaging box, so be careful of their inspection. For special packaging, it is generally after opening the box, all packaging boxes, protective foam, plastic bags should be kept for a period of time. If there is a problem during the return period, you can return or exchange the original. After the return period (usually 10-30 days, according to the manufacturer’s Instruction of Terms of Service), these packaging boxes may be disposed of if there is no problem.